DUPLICATE IMAGE ON THE ENDOCAVITARY TRANSDUCER
Hello guys, I have an HD11XE with the duplicated image only on the endocavitary transducer, can someone help me? does the resintallation of the software solve?
Its a probe problem.it happen in convex probes
most probably it is a probe issue. In some cases power supply also will make similar issue .
test with other probe to narrow it down
And some time the c5-2 probe make mirror and affect on other probe
Yes most it probes in low cases it power supply issue
But you can get another probes to test
replaces the probe and it worked correctly, thank you very much.